- Model NO.: Linvatec QL4300
- Model: Linvatec Ql4300
- Type: Medical Service
Service Circle: 2 working days
Fault: Vaporous image
Solution: Replace one piece rod lens
Remarks: Linvatec QL4300 with vaporous image, after testing, we found one piece of rod lens is broken, and the whole optic system is damaged, then we replace the rod lens, adjust the optic lens until it is back to normal.
Any inquiry for medical equipment repair, repair traning, equipment and parts (pre-owned, used, brandnew, orginal or compatible etc), email or call us directly!