- Model NO.: Eg-2931k
- Model: Pentax Eg-2931k
- Type: Medical Equipment Repair
Service Circle: 2 working days
Fault: No up angle
Solution: Replace angulation wire coil, repair light guide tube
Remarks: After checking, we found the angulation wire coil is broken, and light guide tube is in bad condition, so we did matched repair work. with 90 days warranty after repair.
Any inquiry for medical equipment repair, repair traning, equipment and parts (pre-owned, used, brandnew, orginal or compatible etc), email or call us directly!