Repair High-Frequency Probe for Ge Voluson 730 (SP10-16)
Customer type: Hospital
Equipment name: GE SP10-16 high-frequency probe
Maintenance time: 2010-4-2
Service period: 30 days
Problem Description: Image fuzzy.
Test results: Array of failure.
Solution: Replace lens, crystal, housing and strain relief.
- Guangzhou DODODO Medical Equipment Service Co., Ltd.
- 3rd Year Audited Supplier
- Guangdong, China
- Business Type: Trading Company, Individuals/SOHO, Other
- Main Products: Ultrasound Probe, Endoscope, Medical Repair, Repair Training, Ventilator, Patient Monitor, Video Endoscope Repair, Fiber Endoscope Repair, Medical Accessories, Medical Service
- OEM/ODM Service: No
- Ultrasound (101)
- Ultrasound Repair (14)
- Flexible endoscope (63)
- Flexible (video & fiber )endoscope repair (49)
- Rigid endoscope repair (42)
- Rigid endoscope (20)
- Medical repair training (26)
- Patient Monitor and defibrillator (112)
- Video Camera system (34)
- Ventilator (77)
- Medical lamps (6)
- Medical board repair (4)
- Medical repair tools (2)