Equipment name: GE SP10-16 high-frequency probe
Maintenance time: 2010-4-2
Service period: 30 days
Problem Description: Image fuzzy.
Test results: Array of failure.
Solution: Replace lens, crystal, housing and strain relief.
- Guangzhou DODODO Medical Equipment Service Co., Ltd.
- Audited Supplier
- Guangdong, China
- Business Type: Trading Company, Individuals/SOHO, Other
- Main Products: Ultrasound Probe, Endoscope, Medical Repair, Repair Training, Ventilator, Patient Monitor, Video Endoscope Repair, Fiber Endoscope Repair, Medical Accessories, Medical Service
- OEM/ODM Service: No
- Ultrasound Repair
- Flexible endoscope
- Flexible (video & fiber )endoscope repair
- Rigid endoscope repair
- Rigid endoscope
- Medical repair training
- Patient Monitor and defibrillator
- Video Camera system
- Medical lamps
- Medical board repair
- Medical repair tools