- Guangzhou DODODO Medical Equipment Service Co., Ltd.
- Audited Supplier for 2nd Time
- Guangdong, China Business Type: Trading Company, Individuals/SOHO, Other
- Main Products: Endoscope, Ultrasound Probe, Medical Repair, Repair Training, Ventilator, Patient Monitor, Video Endoscope Repair, Fiber Endoscope Repair, Medical Accessories, Medical Service
- OEM/ODM Serivice: No
- Rigid endoscope repair (24)
- Flexible (video & fiber )endoscope repair (37)
- Ultrasound Repair (14)
- Electrocardiograph (2)
- Medical repair tools (2)
- Surgical instruments (3)
- Dynamical system (11)
- Medical lamps (5)
- Medical repair training (26)
- Flexible endoscope (50)
- Rigid endoscope (20)
- Video Camera system (34)
- Ultrasound (101)
- Patient Monitor and defibrillator (91)
- Ventilator (76)
- Medical board repair (4)
Customer type: Hospital
Equipment name: GE SP10-16 high-frequency probe
Maintenance time: 2010-4-2
Service period: 30 days
Problem Description: Image fuzzy.
Test results: Array of failure.
Solution: Replace lens, crystal, housing and strain relief.