Service Circle:10 working days.
Fault:bending rubber leaking,insertion tube ageing,bending section flatten,biopsy channel crushed,
Vision turned yellow ,down angel steel wire broken.
Solution:Replace insertion tube,bending section and rubber.
Remarks:CCD may increase black spot during disassemble for the snake bone crushed and leakage before.
Any inquiry for medical equipment repair, repair traning, equipment and parts (pre-owned, used, brandnew, orginal or compatible etc), email or call us directly!